The Nonin® Onyx Model Finger Pulse Oximeter is a small, lightweight, The use of accessories other than those specified in these instructions may result . View and Download Nonin Onyx instructions for use online. Finger Pulse Oximeter. Onyx Medical Equipment pdf manual download. Medical Equipment Nonin PalmSAT Operator’s Manual. (35 pages). Medical Equipment Nonin Onyx Instructions For Use. Finger pulse oximeter (5.
|Published (Last):||18 May 2004|
|PDF File Size:||16.68 Mb|
|ePub File Size:||19.31 Mb|
|Price:||Free* [*Free Regsitration Required]|
Indicates separate collection for electrical and electronic equipment WEEE. Do not use this device in an MRI environment, in an explosive atmosphere, or on infant or neonatal patients. Memory Functions m Only 3 minutes after removing the batteries. This device is intended only as an adjunct in patient assessment.
This device should not be used adjacent to or stacked with other equipment. Opening the case may damage the device and void the warranty.
In some circumstances, the device may interpret motion as good pulse quality. Do not autoclave or immerse the device or sensors in iinstruction. This device is designed to determine the percentage of arterial oxygen saturation of functional hemoglobin.
Nonin 8500 Operator’s Manual
Instructuon device is not defibrillation proof per IEC Pilot’s Handbook of Aeronautical Knowledge U. Ensure that the sensor is firmly plugged in.
Any sign or evidence of opening the system, field service by non-Nonin personnel, tampering, or any kind of misuse or abuse of the system, shall void the warranty in its entirety.
Enter text from picture: Due to the very small trace size, extreme care must be used when replacing components to prevent permanent, non-repairable damage to the circuit board. Comments to this Manuals Your Name. Using the and M Pulse Oximeters Figure 2: A flexible circuit connects the two halves. Follow local, state and national governing ordinances and recycling instructions regarding disposal or recycling of the device. Connecting the Sensor Connect the pulse oximeter sensor with the Nonin logo facing up to the top of the device as shown.
For instructions on cleaning pulse oximeter sensors, refer to the respective sensor instructions for use. It is intended for spot-checking of adult and pediatric patients on fingers other than the thumb between 0.
NONIN OPERATOR’S MANUAL Pdf Download.
Do not mix fully charged and partially charged batteries. Memory download function M only. If any indicator is not lit, do not use the device. Page of 33 Go.
Nonin | Diamondback Lode Valiant Treadmill User Manual | Page 38 / 96
This device is intended for use in an electromagnetic environment in which radiated RF disturbances are controlled. By pressing ‘print’ button you will print only current page. Consult Instructions for Use. Remove batteries if the device will be stored for more than.
Technical Information Table 6: Southeastern Equipment carries Tennant floor scrubbers, Advance floor scrubbers, Minutemen floor buffers, many other new and used floor buffers and sweepers, and. The use of accessories noonin than those specified in these instructions may result in increased electromagnetic emission.
Visual Indicators Visual Indicators The following table describes conditions and visual indicators. Installing Batteries in the The ECG monitor may not be Assess the patient. Do not twist or pull the flexible circuit or overextend the device’s spring. Wrist-worn pulse oximeter with bluetooth 950 technology 51 pages. This manual also for: Don’t show me this message again. Technical Information Table 5: Contents Continued Accuracy Testing This device has no audible alarms and is intended only for spot-checking.
Page 5 Figures Figure 1.