The EC help provides an easy access to the information related to the use of Keysight EC Network Analyzer. Pressing Help key on the front panel. EB/EB, please see the Programming Manual. • Installation and Quick Start Guide (Part Number: Ex1, attached to optional ABA). This manual . The EC complies with INSTALLATION CATEGORY II as well as The Service Manual is a guide to servicing the EC ENA Series Network Analyzer .
|Published (Last):||19 June 2004|
|PDF File Size:||2.64 Mb|
|ePub File Size:||17.67 Mb|
|Price:||Free* [*Free Regsitration Required]|
The boot option is displayed. Press the standby switch of the EC to turn it on.
Atec Agilent-EC User Manual | 60 pages
The EC restart automatically. Component replacement and internal adjustments must be made by qualified maintenance personnel.
Network setting GPIB setting Printer setting The driver for the supported printer installed after purchase is deleted. This calibration effectively eliminates the frequency response reflection tracking error, manuwl error, and source match error from the test setup in a reflection test using that port.
Select the saved backup image file, then click Next button. When the recovery is finished, EC restarts. If anything is missing, contact the nearest Agilent Technologies office.
CN or below This section describes how to create the user backup image. This document contains proprietary information that is protected by copyright.
You need to execute initial registration again. It is recommended to save the file to a directory which has enough space like the E drive. A message box appears after the completion of recovery process.
Atec Agilent-E5071C User Manual
Click Done to terminate the 1-port calibration process. The dialog box disappears automatically after a few seconds.
If you perform the following procedure incorrectly, a message asking you whether to return to the previous registration screen and perform the registration appears. Reconfiguring Touch Panel Settings Step 1. Click Port x Open to start the calibration measurement x denotes the test port to which the standard is connected. Dangerous voltage levels, capable of causing f5071c, are present in this instrument.
If you use a calibration kit other than a predefined one, you need to define it.
EC User Manual – EC Network Analyzer from Keysight
Factory recovery Returns the contents of the C drive to the factory s5071c. In this case, follow the instruction to return to the previous registration screen. The firmware revision you installed remains even if you perform factory recovery. The comments can be omitted.
EC firmware is executed, then the measurement display d5071c. Press any key to continue. Select a test port and corresponding S parameter on which 1-port calibration will be performed.
Click OK to proceed. This calibration effectively eliminates the directivity error, crosstalk, source match error, frequency response reflection tracking error, and frequency response transmission tracking error from the test setup in a transmission or reflection test using those ports. All rights are reserved. The manual f5071c number changes when extensive technical changes maanual incorporated. A confirmation dialog box is displayed.
You cannot use the front panel keys during the initial registration of the EC therefore connect the mouse and the keyboard before turning on the power.
After restart, the screen for initial registration appears.
Out-position of push-button switch. Turn ON the EC.